- Can I take measures to promote a good radiation protection practice in dentistry?
- Should patients and carers wear lead aprons and personal protective devices during a dental
- What are the recommendations and safety measures for children
- How can I avoid unnecessary examinations?
- What advice is there to help me select X ray examinations?
- Is it a good idea to monitor the dental development of children using a panoramic
Optimization and radiographic quality
- Should the manufacturer of the X ray set provide me with anything to help with routine QA
checks of X ray equipment?
- What are the most important features of dental X ray examinations that contribute to dose
- How does a digital image receptor affect patient dose in dental radiology?
- How do I know if my radiograph is of a good standard?
- How can I ensure that I get high quality intraoral radiographs?
- How can I ensure that I get high quality panoramic radiographs?
- How can I ensure that I get high quality cephalometric radiographs?
- Can I expect an increased number of rejected films when switching from film to a digital
imaging receptor for intraoral radiography?
- I suspect that my intra oral X ray equipment delivers very high patient doses. Can I do
anything to immediately reduce doses while still obtaining good quality images?
1. Can I take measures to promote a good radiation protection practice
The short answer is YES. The most effective way to reduce dose in dental radiography is to avoid unnecessary X ray
examinations by justification. Routine dental X ray examination for all patients
is not justified In addition, the patient dose for each X ray examination should be optimized so that it is As Low As
Reasonably Achievable (ALARA) and consistent with producing the required image quality. It is important that the
equipment is subject to formal acceptance testing, routine quality control (Figure 5) undergoes proper maintenance, and
has all the standard dose reduction features. Please refer here for details.
Figure 5 scratched CR plate, unsuitable for use
2. Should patients and carers wear lead aprons and personal protective devices during a
dental radiographic procedure?
With well designed and optimized equipment and procedures there is no need for routine use of lead aprons for the
patient in dental radiology. Lead aprons may provide some protection in the case of the vertex occlusal examination,
and may be prudent in the case of vertex occlusal examinations in a patient who is, or may be, pregnant. On the other
hand, the use of a lead apron may reassure patients that every effort is being made to ensure their safety, and may
reduce the amount of time that needs to be taken to reassure them. Certainly a lead apron should be provided for any
patient who requests one. It may also be advisable to consider using them on a cautionary basis where equipment and/or
technique have not been verified by a radiation protection specialist, and where they will not otherwise interfere with
the examination. Thyroid collars should be used in the few examinations where the thyroid may be in the main beam.
Lead aprons should be provided for a person who is required to support a patient during the radiographic procedure
(i.e., a comforter or carer). Assisting adults should be positioned so that all parts of their body are out of the main
3. What are the recommendations and safety measures for children undergoing dental
Many actions are similar to those recommended in adult procedures. Please refer to Question
8 for further details. [AAPD 2008].
Although radiation exposure arising from dental radiology is low, a child may undergo many repeated procedures
during childhood and adolescence. Therefore, the accumulated effect of the radiation exposure should be taken into
consideration. The salivary and the thyroid glands are among the organs at risk in dental radiology. The salivaries are
often within the primary beam, while the thyroid receives dose mainly due to scattered radiation [LOOE, H.K. et al., 2006]. Since the thyroid is one of the most radiosensitive organs in children, it may
be necessary to consider shielding it from time to time; useful guidance in this regard is available in [EC-RP 136].
4. How can I avoid unnecessary examinations?
By making sure that radiographs are selected for each individual patient based on clinical need. You should avoid
using a “routine” protocol for X rays of patients and always examine the patient before choosing any X ray procedures.
Just as you prescribe drug therapy, such as antibiotics or painkillers, to suit a patient’s diagnosis, so you
should try to select any X ray examinations according to their clinical need.
5. What advice is there to help me select X ray
Guidelines have been developed by various professional organizations [EC-RP136; Haute Autorité de Santé; SEDENTEXCT Provisional Guidelines; Espelid et
al., 2003; Harris et al., 2002; Isaacson et al., 2008; Pendlebury et al., 2004] that can support you and the patient in this decision, called referral criteria
(or selection criteria). These are systematically developed statements of “good practice” for radiology in specific
clinical dental situations. They are not rules, but give you a framework against which the needs of your patient can be
6. Is it a good idea to monitor the dental development of children using a panoramic
No, there is no justification for this routine practice. Radiography may be required when a clinical examination
suggests the presence of an abnormality, or when interceptive and active orthodontic treatment is being considered.
Clinical indicators, used to identify patients who might benefit from a panoramic radiograph, are effective in
excluding children for whom an X ray examination is not likely to be of value.
Optimization and radiographic quality
7. Should the manufacturer of the X ray set provide me with anything to help with routine
QA checks of X ray equipment?
Yes. Professional societies, in collaboration with national authorities often recommend that users make regular
image quality performance checks on X ray equipment (and viewing screens where relevant). This is particularly
important for dental cone-beam CT systems and panoramic X ray equipment. To enable users to do this, manufacturers
should provide details of the test procedures and the expected results in the equipment’s instruction manual. Any test
objects or phantoms that are necessary for these tests and specific to individual equipment models or manufacturers
should be provided with the equipment as standard.
8. What are the most important features of dental X ray
examinations that contribute to dose reduction?
Professional societies, in collaboration with national authorities should publish guidance for users of dental X ray
equipment on how to optimize the radiation exposure of patients during justified X ray examinations. For each imaging
modality, there are many actions that can be taken to achieve a significant reduction in dose. These are listed below
for intra oral; panoramic and cephalometric, dental CBCT. In addition, ensuring high quality clinical images a significant means of protecting
patients by maximizing the benefits of the X ray examination [see Question10, 11, 12 and 13].
For intra oral equipment:
- Rectangular collimation which approximates the size and shape of the receptor reduces dose significantly in
comparison to circular collimation; a dose reduction exceeding 60 % can be achieved in dental radiology by using
- The fastest available film consistent with achieving satisfactory diagnostic results should be used. E-speed and
F-speed films reduce dose by more than 50% compared with D-speed films.
- Digital detectors have the potential for further dose reduction, even compared with F-speed film, provided the
repeat rate and use of higher exposure factors than necessary are controlled.
- Using tube voltage in the range 60 to 70 kV.
- The X ray tube filtration should be sufficient to reduce entrance skin dose to the patient consistent with
producing satisfactory image quality.
- A position indication device which ensures a minimum focus-to-skin distance of 20 cm should be attached to the tube
head (eg. by use of a long collimator/cone as opposed to a short conical one).
- Exposure settings used should be the minimum consistent with the speed of the imaging system used. Advice on
exposure settings should be provided in the manual for the X ray equipment, which should be available in the user’s
native language and written in easily understood terminology.
- Where old X ray equipment is used, it may be possible to take immediate action to achieve a significant reduction
in patient dose: [see Question 15].
For panoramic and cephalometric equipment:
- Only the fastest screen-film combinations (at least 400) that are compatible with imaging requirements should be
used for panoramic and cephalometric imaging.Note that the intensifying screen and film must be spectrally matched, for
example, if the screen emits light in the green region of the spectrum, the film used should be one that is sensitive
to green light. Furthermore, the physical condition of screens deteriorates over time and it is important that their
condition is monitored and that badly damaged screens are replaced.
- The X ray beam for cephalometric imaging should be collimated to the area of clinical interest.
- The inclusion of wedge filters in cephalometric equipment reduces exposure to the soft-tissue facial profile and
allows optimal imaging, while the provision of asymmetric collimation allows the exposed area to be confined to the
area of clinical interest.
- Modern panoramic systems also allow the field to be limited to the area of clinical interest, thereby offering a
significant potential for dose reduction. If available, limitation of field size to the area required for diagnosis
should be used for panoramic radiography.
- Where available, paediatric examination modes should always be used for examinations of children. If not available,
the exposure factors (such as kV, mA, exposure time) should be suitably adjusted. This may result in a dose saving to
the patient of 50% or more [Lecomber et al..1993].
For dental CBCT equipment:
- Volume of patient imaged. The smallest volume compatible with the clinical situation should be used where this
provides lower radiation dose to patients.
- kiloVoltage and mAs. These should be optimized for each clinical application and patient
- Choice of voxel size. Some CBCT systems offer a choice of “resolution”. The voxel size compatible with the clinical
situation should be used where this provides lower radiation dose to the patient.
- Number of projections and reconstruction algorithm. Some CBCT systems allow the operator to opt for imaging based
on a reduced number of basis projections. Such options should be used where the resulting image quality is acceptable
for the clinical situation.
When considering buying a CBCT unit, you should check to see whether it is able to comply with national reference
doses for dental CBCT where available .
9. How does a digital image receptor affect patient dose in dental radiology?
- Two types of digital system are used in intraoral, panoramic and cephalometric imaging. One involves imaging
sensors based on charge-couple devices (CCD) and another uses photo-stimulable storage phosphor (PSP) plates (Figure
- Radiographic technique for digital imaging should be adjusted for the minimum patient doses required to provide the
required image quality for each examination type.
- Intraoral digital radiography offers a potential for significant dose reduction; some studies report that,
depending on the diagnostic task, a lower exposure may be used when density and contrast is adjusted using the software
features [EC-RP 136]. This is one of the benefits of digital radiography where image quality can be
optimized after the image has been taken.
- Although digital radiography offers possibility of significant dose reduction, it can, in practice, lead to
increased patient dose. This can arise from, for example: using an image quality higher than is necessary; use of
unduly long exposure times; retakes by staff (e.g. due to bad positioning) that may go undetected; and lack of concern
for collimation. Furthermore, due to smaller sensor size, more than one exposure may be required to cover the
anatomical area imaged using a single conventional film.
Figure 6: a PSP in its plastic cover
If a patient is exposed to X rays for the purpose of producing a radiograph, but the resulting image is not of
adequate quality for clinical use, then the patient has been put at risk for no benefit. Ensuring adequate quality is,
therefore, a fundamental part of radiation protection.
Figures 1 and 2 show excellent quality intraoral and panoramic radiographs.
10. How do I know if my radiograph is of a good standard?
You need to compare your performance by reviewing your radiographic quality against a recognized standard. Such
quality standards for clinical images, and guidance on the audit process, are available [EC-RP
136]. As a minimum target, your aim should be to ensure that no greater than 10% of radiographs are of
unacceptable quality. If you fail this test, then actions can be taken to reduce the proportion of unacceptable
radiographs, with a target of a 50% reduction at each successive audit cycle.
11. How can I ensure that I get high quality
Choosing the correct exposure factors, ensuring accurate patient and X ray source position (using film holders),
along with careful processing should together contribute to achieving excellent results in radiography.
Figure 7a:underexposed intra-oral film
Figure 7b: overexposed intra-oral film
Figure 7c: blurred image
Figure 7d: partial image
For intraoral radiography a simple test tool – an image of a step wedge is useful for maintaining high image quality
(Fig.8) . During installation, a reference standard radiograph of the step wedge should be made using the optimized
exposure setting for an adult/child. Subsequent radiographs of the step wedge/phantom should be made during clinical
use and compared with the reference one to ensure that image quality is maintained [EC-RP 136]
Figure 8: sensitometric steps
12. How can I ensure that I get high quality panoramic
By achieving accurate patient positioning and by good processing of the film. These are the two commonest causes of
poor panoramic radiographic quality. Accurate positioning is helped by using all the positioning aids correctly and by
adequate training. Test tools for panoramic radiography are available [see Question 5 and
13. How can I ensure that I get high quality cephalometric
By using a cephalostat and a fixed X ray source/patient/image receptor relationship. This is achieved using a
dedicated cephalometric attachment to panoramic X ray equipment. In cases where there is no alternative to using a
dental X ray set as the source, it is very important to to ensure correct collimation of the beam and alignment with
14. Can I expect an increased number of rejected films when switching from film to a
digital imaging receptor for intraoral radiography?
Yes, this may happen initially. When switching to a digital image receptor, the retake rate can increase, mainly due
to wrong positioning of the X ray tube and small image receptor with respect to region of interest (ROI). Furthermore,
repeating the exposure is much easier when using digital receptors and this has been reported to lead to increased
reject rates. Careful positioning using sensor-holders with a beam-aiming device and audit of clinical image quality
[see Question 8] will avoid and or reduce retakes.
15. I suspect that my intra oral X ray equipment delivers very high
patient doses. Can I do anything to immediately reduce doses while still obtaining good quality images?
Yes. Older models are more likely to operate below 60 kV, either by design or due to deterioration of the X ray tube
head over its working life. Older models are also more likely to have low values of total filtration. Both low
operating potential (kV) and low filtration are strongly associated with high patient doses, as is the use of speed
group D film [see Question 11], which is often observed to be used with older X
ray equipment. Therefore, an immediate saving in dose can be achieved by taking the following steps, pending the future
replacement of the X ray equipment:
- Move to the use of E-speed film.
- To improve the effective X ray beam quality and provide a lower radiation output rate consistent with the use of
E-speed film, a further 1.0 mm of aluminium beam filtration should be added to the X ray tube head, as close as
possible to the X ray beam window in the tube head. This may require the help of a technician.
- Continue to use the exposure settings you were using before, unless image quality is severely affected. In this
case, the help of a medical physics expert should be sought. The above steps should provide a reduction in dose of at
- Poor film processing conditions may have as great an impact on patient doses as the X ray equipment, and so
attention should also be paid to ensuring that all aspects of processing are carried out in accordance with the advice
provided in [EC-RP 136] and that proper quality assurance methods are in place.
- UNITED NATIONS scientific committee on the
effects of atomic radiation, Sources and effects of ionizing radiation, UNSCEAR Report Volume 1, New York, USA
- INTERNATIONAL ATOMIC
ENERGY AGENCY, Radiological protection for medical exposure to ionizing radiation, Safety Guide, RS-G-1.5, IAEA, Vienna
Commission, European Guidelines on Radiation Protection in Dental Radiology, RP 136, Luxembourg (2004).
Haute Autorité de Santé. Guide des indications et des procedures des examens radiologiques en odontostomatologie
(2006). (accessed 29 August 2009)
- SEDENTEXCT. Radiation Protection; Cone Beam CT for Dental and Maxillofacial Radiology. Provisional
Guidelines (v1.1 May 2009). (accessed 18/8/09))
- Espelid, I., Mejàre, I., Weerheijm, K., EAPD., European Association of Paediatric Dentistry
guidelines for the use of radiographs in children. European Journal of Paediatric Dentistry 4 (2003) 40-48.
- Harris, D., Buser, D., Dula, K., Gröndahl, K., Jacobs, R., Lekholm, U., Nakielny, R., van,
Steenberghe, D., van, der, Stelt, P., E.A.O. Guidelines for the use of diagnostic imaging in implant dentistry.
Clinical Oral Implants Research 13 (2002) 566-570.
- Isaacson, K.G., Thom, A.R., Horner, K., Whaites, E., Orthodontic Radiographs. Guidelines for the use
of radiographs in orthodontics. British Orthodontic Society: London, (2008).
- Pendlebury, M.E., Horner, K., Eaton, K.A., (eds). Selection Criteria for Dental Radiography.
Faculty of General Dental Practitioners (UK) Royal College of Surgeons of England, London, (2004).
- Lecomber, A.R., Faulkner, k., Dose reduction in panoramic radiography. Dentomaxillofac Radiol.
- American Academy on Pediatric Dentistry ad hoc Committee on Pedodontic Radiology:
American Academy on Pediatric Dentistry Council on Clinical Affairs. Guideline on prescribing dental radiographs for
infants, children, adolescents, and persons with special health care needs. Pediatric Dentistry 30 (7 Suppl): 236-7
- LOOE, H.K. et al., Radiation exposure to children in intraoral dental radiology, Rad. Prot. Dosim.
121 (2006) 461-465.
- WAGNER et al. Exposure of the pregnant patient to diagnostic radiations: A Guide to Medical
Management, 2nd Edition, Publisher: Medical Physics Publishing, Madison, (WI 1997).
- NATIONAL RADIOLOGICAL PROTECTION BOARD, Radiation exposure of the UK population, Review NRPB
R263, Chilton, UK (1993).
- NATIONAL COMMISSION FOR RADIATION PROTECTION, Implementation of the principle as low as reasonable
achievable (ALARA) for medical and dental personnel, NCRP Report 107, NCRP (1990).
- AUSTRALIAN RADIATION PROTECTION AND
NUCLEAR SAFETY AGENCY, Radiation Protection in Dentistry, Code of Practice and Safety Guide, RPS 10, Canberra,